10" color touchscreen: 1805 & 1886 series
Modern digital technology for safety-current-limited applications or those at 500 VA / 200 mA.
Power strips must be tested for safety and function following their production.
Industry: electrical industry
It was necessary to develop a testing system to automatically test the power strips with different numbers of sockets, which are positioned differently and also have different functions (e.g. master-slave).
In order to test the large number of different DUT, a test cover was used, in which up to twelve socket adapters with interchangeable head can be used for different types of connectors and can be flexibly positioned. It is also possible to integrate an actuator for the power strip switch. The test cover was developed and supplied by the customer.
After closing the test cover, automatic contact is established between one and twelve sockets via pneumatic cylinders. During the test, the switch integrated in the power strip can be automatically actuated via two additional cylinders to check the function of the switch. The ground bond conductors of all sockets, the high dielectric strength and function are tested. Moreover, a load is connected to the first socket of master-slave power strips to check the switching on of additional sockets.
The easy-to-use PC software allows storage of an arbitrary number of test programs, so that the necessary scope of testing and the test data can be adjusted for each DUT. However, for products with similarly mounted DUT, it is also possible to use the same test program, which is then allocated to the respective DUT over the product list. By coding in different parts of the DUT contact, it is checked with the software whether the settings of the test cover match with the selected test program. The test results are automatically stored in XML or Access format.
A separate test dummy is used to check the functionality of the testing system. The testing system does not allow further testing without a passed dummy test.